Positive Displacement Oval Gear Sensor contains a pair of oval gear rotors. Positive Displacement oval gear sensors where the liquid passage causes two oval gear rotors to rotate within a precious measuring chamber and with each of rotation a fixed volume of liquid is displaced passing through the sensor. Magnets which are inserted in the rotors transfer pulse train output.
Working principle : Oval Gear Positive Displacement
Accuracy (standard installation position) : ±0.5 % FSD
Repeatability : 0.03 %
Maximum working pressure : 2 MPa = 20 bar
Fluid &Ambient temperature : -40 to100°C
Viscosity : 1000 cps maximum
Material of construction :
Rotor and casing : CAST IRON
O-ring : NBR
Shaft : Stainless Steel
Process connection : Standard flange
Flow range : As on bellow model no wise table
Major Fluids : aluminum wetted part : Kerosene, Gasoline, diesel, light Oil, heavy oil
Pink printed dress, woven, round neck with a keyhole and buttoned closure at the back, sleeveless, concealed zip up at left side seam, belt loops along waist with slight gathers beneath, brand appliqu?? above left front hem, has an attached lining.Comes with a white, slim synthetic belt that has a tang clasp.